http___www.trox.hr_stru_C4_8Dnost_projektiranje-dcb5fcd39eaa68ba_1.xml
1.1 2.1 3.1 4.1 5.1 6.1 7.1 8.1 9.1 10.1 11.1 12.1 13.1 14.1 14.2 15.1 16.1 19.1 20.1 20.2 21.1 22.1 23.1 24.1 25.1 26.1 27.1 28.1 28.2 29.1 30.1 31.1 32.1 33.1 34.1 35.1 36.1 37.1 38.1 39.1 40.1 41.1 42.1 43.1 44.1 45.1 45.2 46.1 47.1 48.1 49.1 50.1 51.1 52.1 53.1 54.1 56.1 57.1 58.1 59.1 60.1 62.1 63.1 64.1 66.1 67.1 68.1 69.1 70.1
Electronic, self-powered monitoring system for fume cupboards Expansion options Application Equipment functions Special characteristics Variants Parts and characteristics Attachments Expansion modules are factory mounted or can be fitted at a later stage Optional transducers for FMS-2 Construction features Materials and surfaces Functional description For fume cupboard monitoring, the electronic EASYLAB monitoring system FMS is typically installed on the roof or on the side of the fume cupboard. The supplied control panel is typically mounted on the fume cupboard side frame. Principal functions: Monitoring strategies: Special functions Electronic monitoring system for the demand-based monitoring of differential pressures, volume flow rates or face velocities in fume cupboards; function display to EN 14175 with optical and acoustic signalling. Monitoring strategies: Equipment function FMS-1 Monitoring system with integral static differential pressure transducer, for the measurement and monitoring of two differential pressure values or volume flow rates Equipment function FMS-2 Monitoring system for the connection of two external transducers using a 0 – 10 V DC signal for the measurement and monitoring of two differential pressure, volume flow rate or face velocity values. Special characteristics Materials and surfaces Technical data This specification text describes the general properties of the product. Texts for variants can be generated with our Easy Product Finder design programme. Order example: FMS-1/SG Inputs 1 analog input 2 volt-free digital inputs Outputs 1 analog output 4 digital outputs Switching outputs are provided for activating supportive flow, switching between volume flow rate 1 and 2, alarm signalling Interfaces for external signalling The following information is available with conventional interfaces: Status display LEDs for status information and to signal errors: alarm, supply voltage with heartbeat Additional status LEDs on the PCB: Display of the switching status of the digital inputs and outputs Electrical connection For details on the connecting cable core identification refer to the FMS installation and operating manual. Installation and commissioning The maximum power required depends on the construction of the monitoring system. Typical constructions with different equipment result in the following values: Share page Recommend this page Recommend this page by sending a link by mail. Share page Thank you for your recommendation! Your recommendation has been sent and should arrive shortly. Contact We are here for you Please specify your message and type of request Contact Thank you for your message! Your message is send and will be processed shortly. Our department for Service-Requests will contact you asap. For general question regarding products or services you can also call: Contact We are here for you Please specify your message and type of request Contact Thank you for your message! Your message is send and will be processed shortly. Our department for Service-Requests will contact you asap. For general question regarding products or services you can also call:
1.1 2.1 3.1 4.1 4.2 4.3 5.1 6.1 7.1 8.1 9.1 10.1 11.1 12.1 13.1 14.1 15.1 16.1 17.1 18.1 19.1 20.1 21.1 22.1 23.1 24.1 25.1 26.1 27.1 28.1 30.1 31.1 32.1 33.1 34.1 36.1 37.1 38.1 40.1 41.1 42.1 43.1 44.1
Jednostavno, precizno i učinkovito Tri optimirane kontrolne jedinice odlikuju se najvišom kvalitetom, jednostavnim instaliranjem i preciznim radom. Projektiranje tehničkih usluga u građevinarstvu složen je zadatak. Očekivanja vlasnika često su jednaka: ekološka i ekonomska održivost uz istodobno visoku udobnost i male troškove ulaganja. Ne postoje međutim dva potpuno jednaka objekta ‑ različiti tipovi građevina i njihovo korištenje, bilo da se radi o novoj gradnji ili obnovi, zahtijevaju individualne koncepte. Pronađite najbolja klimatizacijska i ventilacijska rješenja za svoje objekte! Ovdje možete saznati više o sljedećim temama: U kupovini Uredi Laboratoriji Hoteli Zračne luke Škole, centri za dnevnu skrb i druge obrazovne ustanove Bolnice Stadioni Muzeji Čisti zrak Naši priručnici za projektiranje pomažu vam u individualiziranom projektiranju klimatizacijskih sustava. Uz pomoć kriterija projektiranja, zakonskih propisa i standarda, objašnjenja o korištenju pojedinih proizvoda, studija ekonomske izvodljivosti, arhitektonskih mogućnosti i detaljnih pregleda proizvoda možete odabrati komponente, uređaje i sustave koji najbolje odgovaraju vašem projektu. Priručnik za projektiranje AIRFLOWCONTROL Priručnik za projektiranje sustava zrak-voda Kontakt Tu smo za vas Preporučite ovu stranicu slanjem poveznice poštom. Podijeli stranicu Zahvaljujemo na preporuci! Vaša je preporuka poslana i trebala bi stići uskoro. Kontakt Tu smo za vas Navedite svoju poruku i vrstu zahtjeva Kontakt Zahvaljujemo na vašoj poruci! Vaša je poruka poslana i ubrzo će biti obrađena. Naš odjel za servisne zahtjeve što skorije će vas kontaktirati. Za opća pitanja u vezi s proizvodima ili uslugama možete i nazvati: Kontakt Tu smo za vas Navedite svoju poruku i vrstu zahtjeva Kontakt Zahvaljujemo na vašoj poruci! Vaša je poruka poslana i ubrzo će biti obrađena. Naš odjel za servisne zahtjeve što skorije će vas kontaktirati. Za opća pitanja u vezi s proizvodima ili uslugama možete i nazvati: