http___q-lab.com_tr-tr_products_q-sun-xenon-arc-test-chambers_q-sun-xe-3_1.xml
1.1 2.1 2.2 2.3 3.1 4.1 4.2 4.3 4.4 4.5 5.1 6.1 6.2 6.3 6.4 6.5 7.1 8.1 8.2 8.3 8.4 9.1 10.1 10.2 10.3 10.4 11.1 12.1 12.2 12.3 12.4 12.5 13.1 14.1 14.2 14.3 14.4 15.1 16.1 16.2 16.3 16.4 16.5 17.1 17.2 17.3 18.1 18.2 18.3
Full Spectrum Xenon Lamps Xenon arc lamps give the best reproduction of full spectrum natural sunlight. Q-Lab's lamps are air-cooled (not water-cooled), which makes them more economical, highly efficient and very low maintenance. Lamps last for 1500 hours and the Xe-1 uses one lamp per machine. Optical Filters A choice of optical filters is available to simulate a variety of service environments including direct sunlight and sunlight through window glass. The performance of Q-Lab's filters does not decline with age; therefore, filters do not need to be replaced under normal operation. The only exception to this is the Window-IR filter, which requires regular replacement. All Q-SUN Xe-1 models use one optical filter per machine. See LX-5060 - A Choice of Filters for Q-SUN. Temperature Monitoring & Control Control of temperature is important because it influences the rate of degradation. Specimen exposure temperature is precisely controlled in the Q-SUN Xe-1 tester using a black panel temperature sensor. The temperature can be set at any point between 35ºC and 103ºC, depending upon whether the sensor is insulated or uninsulated, irradiance level, filter type, lamp age and ambient room temperature. Black panel temperature control is standard on the Xe-1 tester, or you can substitute chamber air temperature control as an option. See LX-5000 - Q-SUN Xe-1, Xe-2 & Xe-3 Brochure. Versatile Specimen Mounting The flat specimen mounting tray in the Xe-1 tester accommodates different shapes and sizes of specimens, including three-dimensional parts. A variety of specimen holders are available. The Xe-1 mounting tray is 251mm x 457mm and holds up to 17 specimens @ 51mm x 102mm each. See our specimen mounting page, LX-5080-SM - Q-SUN Xe-1 & Xe-3 Specimen Mounting or Accessories Tab for more information. SOLAR EYE Irradiance Control The SOLAR EYE irradiance control system constantly monitors and controls lamp output. This assures precise light exposure and maximizes reproducibility of test results. Irradiance control is available at 340nm, 420nm or TUV (Total UV). See LX-5000 - Q-SUN Xe-1, Xe-2 & Xe-3 Brochure. Optional Water Spray (Models Xe-1-S and Xe-1-SC) Outdoor moisture attack is simulated via an optional pure water spray in the Xe-1 tester. Two spray nozzles are located in the top of the chamber. Spray can be programmed to operate during either the dark or light cycle. Note that purified water is required for proper operation of the Xe-1 with water spray. See LX-5000 - Q-SUN Xe-1, Xe-2 & Xe-3 Brochure. Optional Heated Water Immersion (Model Xe-1-W) The ponding tray in the Q-SUN Xe-1-W submerges test samples in a temperature-controlled water bath during light or dark cycles. Immersion, draining, and temperature control are performed automatically with an integral water repurification and recirculation system. Immersion weathering testing simulates environments experienced by materials frequently exposed to standing water, like roofing materials and photovoltaic panels. See LX-5019, Xe-1-W Product Summary. Optional Chiller for Low Temperature Applications (Models Xe-1-BC and Xe-1-SC) For some interior products such as pharmaceuticals and cosmetics, exposure at room temperature is necessary to prevent unnatural degradation. An optional chiller is available for the Xe-1 tester to lower temperatures when testing such heat sensitive materials. See LX-5000 - Q-SUN Xe-1, Xe-2 & Xe-3 Brochure. With a chiller, a Q-SUN Xe-1 tester can operate at 19-103ºC, depending upon sensor type, irradiance level, filter type, lamp age, and ambient room temperature. The Xe-1 with chiller is configured so that the chiller acts as a permanent stand for the compact tester. Controls & Optional SoftwareFor quality systems that require documented proof of test conditions, every Q-SUN Xe-1 tester is equipped with an Ethernet connection. Optional “Virtual Strip Chart” PC software allows the user to automatically record exposure conditions and directly transfer data from the Q-SUN tester to a Windows-based computer. This data can also be emailed directly to Q-Lab's technical support desk for expert troubleshooting and diagnostics. Designed to be both functional and easy to use, the Q-SUN Xe-1 controller can be programmed in five user-selectable languages (English, French, Spanish, Italian or German) and includes complete self-diagnostic error checking. The controller constantly monitors the status and performance of all systems. It also displays simple warning messages, routine service reminders or performs safety shutdown, as needed.
1.1 2.1 3.1 4.1 5.1 6.1 7.1 8.1 9.1 10.1 11.1 12.1 13.1 14.1 15.1 16.1 17.1 18.1 19.1 20.1 21.1 22.1 23.1
Tam Spektrum Xenon Lambaları Doğal güneş ışığının tam spektrumunun oluşturulmasında Xenon arc lambalar en iyi sonucu vermektedir. Q-Lab'ın lambalarını çok daha ekonomik, çok daha verimli ve de çok daha düşük bakım masraflı yapan özelliği hava soğutmalı (su soğutmalı değil) olmasıdır. Lambaların kullanım ömrü 1500 saattir ve Xe-3 cihaz başına üç lamba kullanır. Optik Filtreler Doğrudan güneş ışığı ve pencere camından gelen güneş ışığı da dâhil olmak üzere çeşitli ortamları simüle etmek için aralarından seçim yapabileceğiniz optik filtreler vardır. Q-Lab filtrelerinin performansı zamanla azalmaz bu nedenle normal işlemler haricinde değiştirilmelerine gerek yoktur. Tek istisna Window-IR filtresi içindir ki bu filtrenin düzenli bir şekilde değiştirilmesi gerekir. Tüm Q-SUN Xe-3 modellerinde cihaz başına üç filtre kullanılmaktadır. Detaylı bilgi için; LX-5060 - Q-SUN için Filtre Seçimi sayfasına bakınız. Sıcaklık İzleme ve Kontrol Sıcaklık kontrolü önemlidir zira bozulma hızını etkiler. Isıya maruz kalan numuneler, siyah panel sıcaklık sensörü kullanılarak Q-SUN Xe-3 test cihazında hassas şekilde kontrol edilir. Sensörün yalıtımlı ya da yalıtımsız olması, ışık seviyesi, filtre tipi, lamba yaşı ve ortam oda sıcaklığına bağlı olarak sıcaklık 42ºC ve 120ºC arasında herhangi bir değere ayarlanabilir. Opsiyonel olarak sağlanan çiller ile numune sıcaklığında üstün bir kontrol sağlanarak hem siyah panel hem de kabin hava sıcaklığı eş zamanlı olarak kontrol edilebilir. Detaylı bilgi için; LX-5000 - Q-SUN Xe-1, Xe-2 ve Xe-3 Broşürü’ne bakınız. Çok Çeşitli Numune Yerleşimi Xe-3 test cihazında bulunan düz numune yerleştirme tepsisine üç boyutlu ürünler de dâhil olmak üzere farklı biçim ve boyutlardaki numuneleri yerleştirebilirsiniz. Buna ek olarak çeşitli numune tutucular da mevcuttur. 451mm x 718mm boyutlarındaki Xe-3 numune tepsisine her biri 51mm x 102mm ebatlarında 55 numune yerleştirilebilir. Detaylı bilgi için; numune yerleşim sayfası, LX-5080-SM - Q-SUN Xe-1 ve Xe-3 Numune Yerleşimi sayfalarını ziyaret ediniz ya da Aksesuarlar sekmesine tıklayınız. SOLAR EYE Işık Kontrolü SOLAR EYE ışık kontrol sistemi lamba çıkışını sürekli olarak izler ve kontrol eder. Bu sistem hassas ışık şiddetini garanti ederken test sonuçlarının tekrarlanabilirliğini yükseltir. Işık kontrolü 340nm, 420nm veya TUV (Tüm UV) için kullanılabilir. Detaylı bilgi için; LX-5000 - Q-SUN Xe-1, Xe-2 ve Xe-3 Broşürü’ne bakınız. Nem Kontrolü Bağıl Nem (RH) kontrolü Q-SUN Xe-3 modellerinde standarttır. Bu uygulama sayesinde kullanıcı bağıl nemi, siyah panel sıcaklığını ve kabin hava sıcaklığını eş zamanlı olarak kontrol edebilir, izleyebilir ve görüntüleyebilir. Unutmayınız ki nem kontrolüne sahip Xe-3 test cihazının düzgün bir şekilde çalışabilmesi için arıtılmış su gerekmektedir. Detaylı bilgi için LX-5000 - Q-SUN Xe-1, Xe-2 ve Xe-3 Broşürü’ne bakınız. Opsiyonel Su Püskürtme (Xe-3-HS, Xe-3-DS, Xe-3-BS, Xe-3-HSC modelleri) Dış ortam nemliliği, opsiyonel saf su spreyi ile Xe-3 test cihazında simüle edilebilmektedir. Kabinin üst kısmında iki adet sprey fiskiyesi vardır. Spreyleme karanlık ya da aydınlık döngüde çalışacak şekilde programlanabilmektedir. Unutmayınız ki Xe-1’in su spreyinin düzgün bir şekilde çalışabilmesi için arıtılmış su gerekmektedir. Detaylı bilgi için: LX-5000 - Q-SUN Xe-1, Xe-2 ve Xe-3 Broşürü’ne bakınız. Opsiyonal Arka Sprey (Xe-3-BS) Xe-3 test cihazı için kullanılan Arka Sprey bazı SAE test metodları için gerekli olmaktadır; Bu sprey sayesinde numunelerin hem önüne hem de arkasına eş zamanlı olarak su püskürtülebilmektedir. Unutmayınız ki Xe-3’ün arka sprey ile düzgün çalışabilmesi için arıtılmış su gerekmektedir. Opsiyonel Çift Sprey (Xe-3-DS) Xe-3 test cihazı için kullanılan Çift Sprey sayesinde asit yağmuru ya da sabun çözeltisi gibi ikinci bir sıvı solüsyon test numunelerinin üzerine püskürtülebilmektedir. Harici büyük bir rezervuar, santrifüjlü pompa ve filtre Xe-3 test cihazının hemen yanında duracak olan portatif bir düzeneğe monte edilir. Detaylı bilgi için: LX-5025a – Otomobil için Asit Aşındırması broşürüne bakınız. Düşük Sıcaklık Uygulamaları için Opsiyonel Soğutucu (Xe-3-HSC) İlaç ve kozmetik gibi bazı iç mekân ürünlerinin doğal olmayan bozulmalarını engellemek için oda sıcaklığında tutulmaları gerekmektedir. Bu tip ısıya duyarlı ürünleri test ederken sıcaklığı düşürebilmek için Xe-3 test cihazında kullanabileceğiniz opsiyonel bir soğutucu (çiller) mevcuttur. Detaylı bilgi için; LX-5000 - Q-SUN Xe-1, Xe-2 ve Xe-3 Broşürü’ne bakınız. Çiller ile birlikte Q-SUN Xe-3 test cihazı, sensör tipine, ışık seviyesine, filtre tipine, lamba yaşına, oda sıcaklığına ve neme bağlı olarak 31-120ºC aralığında ve %50 bağıl nem (RH) değerinde çalışabilir. Çillerli Xe-3 test cihazında, çiller ünitesi test cihazı yanında duracak şekilde tasalanmıştır ve kolayca yer değiştirmesini sağlayan tekerleklere sahiptir. Kontrol Ünitesi ve Opsiyonel Yazılım Hem işlevsellik hem de kolay kullanım için tasarlanan Q-SUN Xe-3 kontrol ünitesinde otomatik hata denetim sistemi mevcuttur ve (İtalyanca ya da Almanca, İspanyolca, İngilizce, Fransızca) beş kullanıcı tarafından seçilebilir dilde programlanabilir. Kontrol ünitesi tüm sistemin durum ve performansını sürekli takip eder. Ayrıca basit uyarı mesajlarını ve rutin servis hatırlatmalarını görüntüler ve gerekli olan durumlarda güvenli kapatma işlemini de gerçekleştirebilir. Test şartlarının belgelenmiş ispatını isteyen kalite sistemleri için, her Q-SUN Xe-3 test cihazı bir Ethernet bağlantısına sahiptir. Opsiyonel sunulan “Virtual Strip Chart” PC yazılımı kullanıcının test koşullarını otomatik olarak kaydetmesine ve elde edilen veriyi Q-SUN Xe-1 test cihazından Windows tabanlı bilgisayara doğrudan aktarabilmesine olanak tanır. Ayrıca bu veri, uzman sorun giderme ve teşhis çalışması için doğrudan Q-Lab’ın teknik destek birimine e-posta olarak yollanabilmektedir.