<p>When you look on meteorological maps the "surface pressure" is actually the reduced MSLP (mean sea-level pressure) value.  <a href="https://www.weather.gov/bou/pressure_definitions" rel="nofollow noreferrer">See this NWS page for more info</a>.  So whatever height the sensor is at above the ground, they regardless compare that height back versus the sea-level to report the pressure.</p>

<p>I always figured barometers were sited along with the main instrument set.  Those indeed will all generally be raised up off the ground certain heights (aside from any soil temperature thermometers) to prevent issues with any standing water or snow cover.  See for example the <a href="https://www.mesonet.org/index.php/site/about/instruments" rel="nofollow noreferrer">instrument list</a> and <a href="https://www.mesonet.org/common/meta/mesonet/photos/2013/" rel="nofollow noreferrer">extensive pictures</a> for Oklahoma Mesonet site locations (slightly different setup, but the 2 meter temperature and 10 meter wind are widespread standards).</p>

<p>But I was surprised to read in the <a href="https://www.weather.gov/media/asos/aum-toc.pdf" rel="nofollow noreferrer">ASOS User's Guide</a> that:</p>

<blockquote>
  <p>At virtually all locations, the pressure sensors are located indoors
  within the ACU.</p>
</blockquote>

<p>And:</p>

<blockquote>
  <p>The ACU, which is the central processing unit for the ASOS, is usually
  located inside a climate controlled structure, such as an observing
  office or control tower building. It ingests data from the DCP(s) and
  pressure sensors, and is capable of accepting information from the FAA
  New Generation Runway Visual Range (NGRVR) system.</p>
</blockquote>

<p>So it sounds like the barometer is often not on site, and there are indeed no specifications for height.  But since it's reduced to sea level regardless, the elevation is all that really matters to <a href="https://www.mide.com/pages/air-pressure-at-altitude-calculator" rel="nofollow noreferrer">reduce it to MSLP</a>, and so some variation in height of the sensor shouldn't matter.</p>

CC BY-SA 4.0
2018-05-17T04:36:56.707Z
2018-05-18T01:09:27.147Z
2018-05-18T01:09:27.147Z
JeopardyTempest
<p>In various atmospheric equations, surface pressure is one of the variables used. For example, in calculating the density of air or pressure-corrected relative optical air mass. How is surface pressure measured? </p>

<p>I imagine it's the pressure at the height of a sensor on a weather station. Is that considered effectively pressure at Earth's surface / a good-enough approximation, or do scientists generally adjust measurements from weather stations to be corrected for the height difference between the sensor and Earth's surface? Or is surface pressure measured by a sensor actually at Earth's surface?</p>

<measurements><atmospheric-radiation><barometric-pressure>
How is surface pressure generally measured, and at what actual height?
cr0
# How is surface pressure generally measured, and at what actual height?

(Asked by: username_0 on 2018-05-16)

In various atmospheric equations, surface pressure is one of the variables used. For example, in calculating the density of air or pressure-corrected relative optical air mass. How is surface pressure measured?

I imagine it's the pressure at the height of a sensor on a weather station. Is that considered effectively pressure at Earth's surface / a good-enough approximation, or do scientists generally adjust measurements from weather stations to be corrected for the height difference between the sensor and Earth's surface? Or is surface pressure measured by a sensor actually at Earth's surface?

**Tags:** measurements, atmospheric-radiation, barometric-pressure


---

## 1 answers

### Answer by username_1 on 2018-05-17. Score: 2 | Accepted answer

When you look on meteorological maps the "surface pressure" is actually the reduced MSLP (mean sea-level pressure) value. [See this NWS page for more info](https://www.weather.gov/bou/pressure_definitions). So whatever height the sensor is at above the ground, they regardless compare that height back versus the sea-level to report the pressure.

I always figured barometers were sited along with the main instrument set. Those indeed will all generally be raised up off the ground certain heights (aside from any soil temperature thermometers) to prevent issues with any standing water or snow cover. See for example the [instrument list](https://www.mesonet.org/index.php/site/about/instruments) and [extensive pictures](https://www.mesonet.org/common/meta/mesonet/photos/2013/) for Oklahoma Mesonet site locations (slightly different setup, but the 2 meter temperature and 10 meter wind are widespread standards).

But I was surprised to read in the [ASOS User's Guide](https://www.weather.gov/media/asos/aum-toc.pdf) that:

> At virtually all locations, the pressure sensors are located indoors
> within the ACU.

And:

> The ACU, which is the central processing unit for the ASOS, is usually
> located inside a climate controlled structure, such as an observing
> office or control tower building. It ingests data from the DCP(s) and
> pressure sensors, and is capable of accepting information from the FAA
> New Generation Runway Visual Range (NGRVR) system.

So it sounds like the barometer is often not on site, and there are indeed no specifications for height. But since it's reduced to sea level regardless, the elevation is all that really matters to [reduce it to MSLP](https://www.mide.com/pages/air-pressure-at-altitude-calculator), and so some variation in height of the sensor shouldn't matter.


---

