Schauer JAC0524 Charger: 24 Volt, 5 Amp with XLR Connector
Schauer 24 Volt / 5 Amp Fully Automatic Electronic Smart Charger with 3 Pin XLR DC Plug for Medical Mobility Applications 3-Pin XLR Plug, #1 Pin POS(+) Fully Automatic Electronic Charger Intelligent charger follows battery impedance to end of charge cycle for all battery types. Fully automatic. Red LED indicates AC power on. Yellow LED indicates battery is charging. Green LED indicates float mode. Charge control: modified constant current charge to a constant finishing voltage then to a float standby. Current limiting. Fully automatic 3 stage charger can be left on the battery indefinitely. For any type lead acid battery, including conventional, maintenance free, deep cycle, gelled-type, valve regulated batteries. Finishing voltage: 2.28 volts per cell float; 2.4 volts per cell finishing. May be preset to customer's specification for battery type. Not intended for use as a DC power supply. Full load at 2.1 volts per cell continuous - 100% duty cycle. Short circuit proof. Reverse polarity protection (except 2 amp model). Mountable, aluminum case. 6.5"W x 3.5"H x 1.8"D 1.8 lbs
https://m.media-amazon.com/images/I/514o67fvHCL._AC_SL1000_.jpg https://m.media-amazon.com/images/I/71gnyA+fwkL._AC_SL1100_.jpg https://m.media-amazon.com/images/I/719bDPhQfaL._AC_SL1299_.jpg
https://m.media-amazon.com/images/I/41RKLc3wMQL._AC_.jpg https://m.media-amazon.com/images/I/518sZaf310L._AC_.jpg https://m.media-amazon.com/images/I/515DlFS1bPL._AC_.jpg
https://m.media-amazon.com/images/I/41RKLc3wMQL._AC_US40_.jpg https://m.media-amazon.com/images/I/518sZaf310L._AC_US40_.jpg https://m.media-amazon.com/images/I/515DlFS1bPL._AC_US40_.jpg
MAIN PT02 PT03
Schauer
{"Product Dimensions": "6.5 x 3.5 x 1.8 inches", "Item Weight": "2.44 pounds", "Item model number": "0891-24", "Is Discontinued By Manufacturer": "No", "Date First Available": "September 13, 2013", "Brand": "Schauer", "Input Voltage": "240 Volts", "Current Rating": "5 Amps"}