Mastech Variable Regulated DC Power Supply 30V 10A HY3010E
The MASTECH HY3010E VARIABLE DC POWER SUPPLY has 0 to 30 volts at 0 to 10 AMP.The voltage is variable from 0 to 30 volts and the current is also variable from 0 to 10 amps. The digital LED meters display the actual voltage and current within 2 percent deviation. Both the voltage and the amperage are adjustable with COARSE and FINE tuning controls for BEST precision.The MASTECH HY3010E VARIABLE DC POWER SUPPLY is excellent for multiple regulated power requirements with the variable 0-30 volt 0-10 amp output. Other uses include lab, science projects or any other use where stable current is required. Output Voltage: 0- 30V @ 0-10A Output Current: 0-10 A Input Voltage: 110V 60Hz Constant Voltage/ Constant Current Voltage and Current Variable Power effect: CV<0.01%+1mV, CC<0.2% + 1mA Overload effect: CV<0.01% +3mV, CC<0.2% +3mA Waves and noises: CV<2mV r.m.s, CC <10mA, r.m.s Current indication accuracy: LED±2%+2digits, analogue display 2.5% Voltage indication accuracy: LED±1%+2digits, analogue display 2.5% Protection: constant current and short-circuit protection CV≤0.1%+5mV(I>6A) CC≤0.5%+20mA(I>6A) High current limited Load regulation : CV≤0.02%+10mV(I≤6A) CC≤0.5%+10mA(I≤6A) CV≤0.1%+5mV(I>6A) CC≤0.5%+20mA(I>6A) Ripple & Noise: CV≤3mVr.m.s CC≤10mA r.m.s Digital Voltage and current meters built in with Voltage resolution: 2.5% of display reading; 2.5% of Current resolution: display reading Environmental : 0 - 102 F ( 0 -40 C), relative humidity: <90% Size: 14 1/4" X 10 1/4" X 6 1/2" Weight: 11.5 lbs 6 Months Warranty
https://m.media-amazon.com/images/I/4181LYNMmCL.jpg
https://m.media-amazon.com/images/I/4181LYNMmCL._SX38_SY50_CR,0,0,38,50_.jpg
MAIN
Mastech
{"Manufacturer": "Mastech", "Date First Available": "February 5, 2013"}