More bang for your buck: C-arm
by Lisa Chamoff, Contributing Reporter | March 30, 2015
Rating saved
From the March 2015 issue of HealthCare Business News magazine
Buying a used C-arm? Here's what to look out for
The market for refurbished C-arms, which have a long lifespan, remains steady as hospital budgets are challenged. Matthew Blaustein, president of Bluestone Diagnostics, a New York-based C-arm dealer, thinks that facilities will continue to look at the used market because of the savings that can be accrued through the purchase of pre-owned systems, which run roughly 30 to 40 percent less than new C-arms. He sells many refurbished OEC 9800s and 9900s.
"The used market is pretty robust," Blaustein says. When buying a refurbished C-arm, it's most important to closely examine the "glassware" — the image intensifier and X-ray tube, the parts of a C-arm that are crucial to the operation of a system, and can wear out as the system ages, Blaustein says. Artifacts, which are imperfections in an image intensifier typically caused by hair or dust, can degrade the image quality and should be able to be cleaned.
Times Displayed: 235125
Times Visited: 2165
Servicing GE, Philips and Siemens CT equipment with OEM trained engineers
Numed, a well established company in business since 1975 provides a wide range of service options including time & material service, PM only contracts, full service contracts, labor only contracts & system relocation. Call 800 96 Numed for more info.
"If they don't come off, that really creates a problem," Blaustein says. "Even though some artifacts may be negligible, some physicians may find that they detract from what they are specifically looking at." Also, as C-arms are moved around quite a bit, the image intensifier casing can get dented, which can possibly signify excessive or careless use.
Check cables, particularly the high-voltage cable that sends power to the C-arm, to make sure the casings aren't torn, which could cause safety and power issues. Torn cables require replacement as opposed to repair, and a high-voltage cable for an OEC 9800 runs around $3,000, Blaustein says.
Mark Ardoin, the owner of Omni Imaging Service, says that when he rebuilds C-arms, aside from checking for high voltage cable problems, he makes sure that the cable ties are cut just inside the C and also at the electronics cabinet, which prevents breaks in the cables due to stress.
Ardoin also checks the output plug connectivity of the C-arm. Connectivity issues can cause voltage fluctuations and multiple errors in the C-arm. "Careful inspection prior to purchasing is a must for an older unit," Ardoin says.
Richard Stock, president of Radiological Imaging Services, advises facilities to check the ages of both the image intensifier and the X-ray tube. If the system is controlled by software, check that the software is up to date and find out whether the manufacturer will make software upgrades available for the system and if they will support the current system that the software is on, Stock says. Stock also advises purchasers to check the error log, if there is the ability to do so.
Sign up for Weekly Top stories
March 2015 Magazine
Read more stories from the March 2015 issue
More bang for your buck: Special Procedures
Keeping helium costs and temperatures down
Tips for buying used patient monitors
More bang for your buck: Infusion Pumps
3 tips for getting the most out of your molecular imaging equpiment
Five Minutes in Healthcare - featuring Christopher Malter
Medical errors are still killing patients in the U.S, let's talk about it
When the levee breaks on chronic conditions
Beyond typical RTLS use cases
Four tips for implementing a successful virtual care strategy
Five Minutes in Healthcare - featuring Dr. Sameek Roychowdhury
Clinical data connectivity solution: Offset pandemic impact, improve quality measures and reporting, facilitate shift to value
A sad day for the U.S.