THE SERIES DPMX EXTRA LARGE DIGITAL PANEL METER can be easily viewed from across a room or in dark areas.
The 2.3" (57mm) The large LED display segments are available in red, green, or blue. These panel meters come equipped with a universal power supply and a range of user selectable process inputs which include 4-20mA & 0-10V DC.
Input Impedance:Set Voltage: >1 MN) (>10 MN) on 200 mV range); Adjustable Voltage: 392 kN); Adjustable Current: 300N) nominal.Accuracy:B1(1% F.S. + 1 count).
Power Supply:90 to 250 VAC @ 12 VA or 10.5 to 30 VAC/DC @ 6 VA (depending on model). Display:3-1/2 digits, 2.3" height, 7 segment LEDSampling Rate:3 readings per second.
Set Voltage: +/-200 mVDC, +/-2 VDC, +/-20 VDC; Adjustable Voltage: 200 mVDC, 5 VDC, 0-10V DC; Adjustable Current: 4-20mA DC.
Input Impedance: Set Voltage: >1 MΩ (>10 MOhm on 200 mV range); Adjustable Voltage: 392 kOhm; Adjustable Current: 300 Ohm nominal.
Accuracy: +/-(1% F.S. + 1 count).
Power Supply: 90 to 250 VAC @ 12 VA or 10.5 to 30 VAC/DC @ 6 VA (depending on model).
Sampling Rate: 3 readings per s.
Operating Temperature: 14 to 122 DegF ( -10 to 50 DegC).
Storage Range: -40 to 167 DegF (-40 to 75 DegC).
Mounting: 180 Degree gimbal mounting with 30 Degree stops or bezel mount.