What is Sleep Apnea?
Need Supplementary Oxygen?
CPAP (Continuous Positive Airway Pressure)
APAP (Auto-Adjusting Positive Airway Pressure)
BIPAP / Bi-Level (Inhalation + Exhalation Pressure)
Stationary Oxygen Systems
Portable Oxygen Systems
Suction Units
Sleep Screening
Set Up, Installation and Education
Service, Repairs and Parts
Instructional Guides
Portable oxygen concentrators are medical devices that assist people who have a low level of oxygen in their blood. They are powered by plugging the device into an electrical outlet or by using a battery.
If a battery is used, then it will need to be charged by plugging it into an electrical outlet. Most concentrators also come with an adapter so you can use the device while you drive.
A portable oxygen concentrator receives air, purifies it, and then distributes the newly formed air. Before it goes into the concentrator, air is made up of 80 percent nitrogen and 20 percent oxygen.
After the air goes through the oxygen concentrator, it comes out as 90 to 95 percent pure oxygen and 5 to 10 percent nitrogen.
The nitrogen is separated to give the patient the highest dose of oxygen possible, as it is difficult to get that percentage of oxygen without the help of a medical device.
What Makes Up a Portable Oxygen Concentrator?
There are many parts that make up a portable oxygen concentrator. A compressor and sieve bed filter are a couple of the main parts. The compressor compresses air that is filtered into the concentrator, then delivers the air in a continuous stream. The compressed air moves to the sieve bed filters. The sieve bed filter plays an important role, as it is the device that removes the nitrogen from the air. A material called Zeolite, which is a six-sided microscopic cube with holes on each side, is in the sieve bed and this is what removes the nitrogen from the air. Two sieve beds are located in the concentrator. After air is first compressed in the concentrator, it is forced into the first sieve bed. Oxygen is sent into the product tank. The first sieve bed then gets filled up with nitrogen. Next, the gas flow is switched and the compressed air is moved to the second sieve bed. The first sieve bed's compressor is sent to the outside room and the air from the product tank goes back into the first sieve bed. The drop in pressure from the first sieve bed and the weakening of oxygen makes the Zeolite release nitrogen. The Oxygen and Nitrogen come back together and are released in the room as regular air. The air is then compressed and sent to the second sieve where Oxygen is moved through it to the Product Tank. The whole cycle starts over again with the first sieve after a few seconds. Other important parts are the cooling system that keeps the portable oxygen concentrator from overheating, and the nasal cannula that delivers the purified oxygen after the oxygen has been passed through all the sieve bed filters. The cannula helps improve oxygen absorption.
Is A Portable Oxygen Concentrator Good for Day and Night?
Portable oxygen concentrators have two settings for receiving oxygen: pulse dose and continuous flow. The pulse dose mode is usually used for daytime use, as it delivers air via the cannula when you inhale. Concentrators with pulse dose technology also are more compact in design and offer a longer battery life. The continuous flow mode delivers a constant air flow via the tubes. For people who need oxygen while they sleep, this mode is the best option.
Rebates and Government Assistance
Individuals in their own homes can claim a rebate to offset the electricity costs of running the machine. Further Information on NSW and Vic rebates can be found here, or via contacting Albury Vital Air
© 2022 - Albury Vital Air