:DIGitize
:DIGitize [<source>[,..,<source>]]
<source>
string
CHANnel<n> FUNCtion MATH SBUS{1} FFT ABUS EXT
The source to be digitized. MATH is an alias for FUNCtion. The EXT source is available on 2-channel oscilloscopes only. The <source> parameter may be repeated up to 5 times.
<n>
integer
1
The channel number in NR1 format.
The :DIGitize command is a specialized RUN command. It causes the instrument to acquire waveforms according to the settings of the :ACQuire commands subsystem. When the acquisition is complete, the instrument is stopped. If no argument is given, :DIGitize acquires the channels currently displayed. If no channels are displayed, all channels are acquired. The :DIGitize command is only executed when the :TIMebase:MODE is MAIN or WINDow. To halt a :DIGitize in progress, use the device clear command.
Capture an acquisition using :DIGitize.
myScope.WriteString ":DIGitize CHANnel1"