:TRIGger:HOLDoff
:TRIGger:HOLDoff <holdoff_time>
:TRIGger:HOLDoff?
<holdoff_time>
NR3 format
10
The holdoff time value in seconds
The :TRIGger:HOLDoff command defines the holdoff time value in seconds. Holdoff keeps a trigger from occurring until after a certain amount of time has passed since the last trigger. This feature is valuable when a waveform crosses the trigger level multiple times during one period of the waveform. Without holdoff, the oscilloscope could trigger on each of the crossings, producing a confusing waveform. With holdoff set correctly, the oscilloscope always triggers on the same crossing. The correct holdoff setting is typically slightly less than one period.
<holdoff_time><NL> <holdoff_time> ::= the holdoff time value in seconds in NR3 format.
{"6e-08 is not of type 'integer'",["parameters", "0", "minimum"],"type","integer",6.0000e-8,{"integer","Minimum value for numerical parameters"}}