:CHANnel2:SKEW
:CHANnel2:SKEW <skew value>
:CHANnel2:SKEW?
<skew value>
NR3
100
The skew time in NR3 format
ns
The :CHANnel2:SKEW command sets the skew between channels 1 and 2. The maximum skew is +/-100 ns. You can use the oscilloscope's analog probe skew control to remove cable delay errors between channel 1 and channel 2.
<skew value><NL>
<skew value> ::= skew value in NR3 format
Set the skew between channels 1 and 2
:CHANnel2:SKEW -50ns