:TRIGger:SWEep
:TRIGger:SWEep <sweep>
:TRIGger:SWEep?
<sweep>
string
AUTO NORMal
The trigger sweep mode
The :TRIGger:SWEep command selects the trigger sweep mode. When AUTO sweep mode is selected, a baseline is displayed in the absence of a signal. If a signal is present but the oscilloscope is not triggered, the unsynchronized signal is displayed instead of a baseline. When NORMal sweep mode is selected and no trigger is present, the instrument does not sweep, and the data acquired on the previous trigger remains on the screen.
<sweep><NL> <sweep> ::= current trigger sweep mode
Select the AUTO sweep mode
:TRIGger:SWEep AUTO
Select the NORMAL sweep mode
:TRIGger:SWEep NORM
Query the current trigger sweep mode
:TRIGger:SWEep?