:DVM:MODE
:DVM:MODE <dvm_mode>
:DVM:MODE?
<dvm_mode>
string
ACRMs DC DCRMs FREQuency
The DVM mode to set
The :DVM:MODE command sets the digital voltmenter (DVM) mode:
- ACRMs - displays the root-mean-square value of the acquired data, with the DC component removed.
- DC - displays the DC value of the acquired data.
- DCRMs - displays the root-mean-square value of the acquired data.
- FREQuency - displays the frequency counter measurement. Requires the EDGE or GLITch trigger mode, and the DVM source and the trigger source must be the same analog channel.
<dvm_mode><NL> where <dvm_mode> is one of {ACRM, DC, DCRM, FREQ}
Set the DVM mode to AC RMS
:DVM:MODE ACRMs