:TIMebase:MODE
:TIMebase:MODE <value>
:TIMebase:MODE?
<value>
string
MAIN WINDow XY ROLL
The time base mode to set
The :TIMebase:MODE command sets the current time base. There are four time base modes: MAIN - The normal time base mode is the main time base. It is the default time base mode after the *RST (Reset) command. WINDow - In the WINDow (zoomed or delayed) time base mode, measurements are made in the zoomed time base if possible; otherwise, the measurements are made in the main time base. XY - In the XY mode, the :TIMebase:RANGe, :TIMebase:POSition, and :TIMebase:REFerence commands are not available. No measurements are available in this mode. ROLL - In the ROLL mode, data moves continuously across the display from left to right. The oscilloscope runs continuously and is untriggered. The :TIMebase:REFerence selection changes to RIGHt.
<value><NL>
<value> ::= {MAIN | WIND | XY | ROLL}
Set the time base mode to main
:TIMEBASE:MODE MAIN