:DVM:ARANge
:DVM:ARANge <setting>
:DVM:ARANge?
<setting>
string
OFF 0 ON 1
Turns the digital voltmeter's Auto Range capability on or off
If the selected digital voltmeter (DVM) source channel is not used in oscilloscope triggering, the :DVM:ARANge command turns the digital voltmeter's Auto Range capability on or off. When on, the DVM channel's vertical scale, vertical (ground level) position, and trigger (threshold voltage) level (used for the counter frequency measurement) are automatically adjusted. The Auto Range capability overrides attempted adjustments of the channel's vertical scale and position. When off, you can adjust the channel's vertical scale and position normally.
<setting><NL>
<setting> ::= {0 | 1}
Turn on the digital voltmeter's Auto Range capability
:DVM:ARANge ON
Turn off the digital voltmeter's Auto Range capability
:DVM:ARANge OFF