:EXTernal:RANGe
:EXTernal:RANGe <range>[<suffix>]
:EXTernal:RANGe?
range
number
vertical full-scale range value in NR3 format
suffix
string
V mV
unit of the range value
The :EXTernal:RANGe command sets the external trigger input signal range. When using DSOX1200-Series oscilloscopes, this range is either 1.6 V or 8 V when you are using a 1:1 probe. When using EDUX1000-Series oscilloscopes, this range is 8 V when you are using a 1:1 probe and cannot be changed. If the probe attenuation is changed, the range value is multiplied by the probe attenuation factor.
<range_argument><NL> where <range_argument> is the external trigger range value in NR3 format.
Set the external trigger range to 1.6V
:EXTernal:RANGe 1.6V
Set the external trigger range to 8V
:EXTernal:RANGe 8V