:MARKer:MODE
:MARKer:MODE <mode>
:MARKer:MODE?
<mode>
string
OFF MEASurement MANual WAVeform
Sets the cursors mode
The :MARKer:MODE command sets the cursors mode:
- OFF - removes the cursor information from the display.
- MANual - enables manual placement of the X and Y cursors. If the front-panel cursors are off, or are set to the front-panel Hex or Binary mode, setting :MARKer:MODE MANual will put the cursors in the front-panel Normal mode.
- MEASurement - cursors track the most recent measurement. Setting the mode to MEASurement sets the marker sources (:MARKer:X1Y1source and :MARKer:X2Y2source) to the measurement source (:MEASure:SOURce). Setting the measurement source remotely always sets the marker sources.
- WAVeform - the Y1 cursor tracks the voltage value at the X1 cursor of the waveform specified by the X1Y1source, and the Y2 cursor does the same for the X2 cursor and its X2Y2source.
<mode><NL> <mode> ::= {OFF | MEAS | MAN | WAV}
Set the cursors mode to manual
:MARKer:MODE MANual