:EXTernal:POSition
:EXTernal:POSition <value>
:EXTernal:POSition?
value
number
Ext Trig waveform vertical position in divisions in NR3 format
The :EXTernal:POSition command sets the external trigger input waveform's vertical position on the oscilloscope display. When the external trigger input's waveform is displayed (see :EXTernal:DISPlay), a cyan digital waveform, based on the :EXTernal:LEVel threshold voltage, appears on the oscilloscope display. This waveform is one division tall, and the base of the waveform can be positioned from -3.5 divisions to 2.5 divisions. In the InfiniiVision DSOX1204A/G oscilloscopes, the :EXTernal:POSition command can be set to zero only (since the Ext Trig input cannot be displayed as a digital waveform).
<value><NL>
Set the external trigger input waveform's vertical position
:EXTernal:POSition <value>
Query the external trigger input waveform's vertical position
:EXTernal:POSition?