:MEASure:VTIMe?
:MEASure:VTIMe? <vtime_argument>[,<source>]
:MEASure:VTIMe?
vtime_argument
float
Time from trigger in seconds
source
string
CHANnel<n> FUNCtion MATH WMEMory<r> EXTernal
Waveform source
The :MEASure:VTIMe? query returns the value at a specified time on the source specified with :MEASure:SOURce. The specified time must be on the screen and is referenced to the trigger event. If the optional source parameter is specified, the current source is modified. For the EXT digital waveform source, the returned value is either 1 or 0, based on the threshold voltage setting.
<value><NL> where <value> is the value at the specified time in NR3 format
Query the value at a specified time on the current source
:MEASure:VTIMe? 0.01
Query the value at a specified time on a specific source
:MEASure:VTIMe? 0.01, CHANnel1