1.Spectro Colorimeter Introduction
Colorimeter is also called color meter. It is widely used in
different industries such as plastic cement, printing, paint,
weaving and dyeing.It measures the sample color data L*a*b*,
L*c*h*, color difference ΔE and ΔLab according to CIE color space.
2.Spectro Colorimeter Advantages
1).Spectro colorimeter:The highest precision colorimeter
2).Adopts spectral measurement working theory which greatly improve
3).Patented technology guarantees measurement stability
4).Obtain national metrology certification
6).Mass storage memory:100 pcs targets and 200 pcs samples
3.Spectro Colorimeter Technical Data
|Illumination||di/8(Diffused Illumination, 8 degree viewing)|
(conform to CIE No.15,ISO 7724/1,ASTM E1164,DIN 5033 Teil7, JIS
Z8722 Condition c standard)
|Size of integrating sphere||Φ40mm, Avian diffused reflection surface coating|
|Illumination Light source||CLED|
|Measurement Aperture||11mm,optional 4mm,6mm,15mm|
|Repeatability||Standard Deviation ΔE*ab 0.08(when a white calibration plate is
measured 30 x at 10-second intervals after calibration)|
|Observe Angles||2° and 10°|
|Light Source||A,C,D50 and D65|
|Display||Chromaticity value(L*a*b,L*C*h), delta E value, pass/fail, color
tendency, average, generate test report|
|With camera to see the measurement area|
|Color Difference Formula||ΔE*ab,ΔE*CH|
WI(ASTM E313-10,ASTM E313-73,CIE/ISO,AATCC,Hunter,Taube
YI(ASTM D1925,ASTM E313-00,ASTM E313-73)
|Data Storage||20000 samples|
|Light Source Lifetime||5 years, 1.5 million times|
|Other Function||without||camera view, input color swatches|
|Screen||Panchromatic True Color Screen|
|Language||Chinese and English|
|Operating Temperature||5~45℃, relative humidity 80% or below( at 35°C ),no condensation|
|Storage temperature range||-25℃ to 55℃,relative humidity 80% or below(at 35°C ),no
|Power||Rechargeable Lithium Battery 8.4V/2000mAh, adaptor DC12V|
|Standard Accessories||adapter, operating manual, color management software, drive
software, electronic manual, color management guide, USB cable,
black/white calibration tube, protective cover, portable bag,
electronic color charts|
|Color Matching System||Not matched|
|UV Light Source||without|
4.Spectro Colorimeter Longest Guarantee
a. One Year Guarantee
b. If testing result is not correct, we will refund to you
c.Provide testing report for every device to assure the data
authority which saves calibration cost for 1 year
5.Product Pictures of Spectro Colorimeter
6.Spectro Colorimeter Free Color QC Software
Device is with mature color QC software which can connect the
colorimeter with PC. With the software, we could save and manage
the color data, derive the data, generate testing report,print test
report. It will help the chemist finish the job much easier.
(Software english version is also available.)
Packaging & Shipping
Packing Material:canvas bag with carton box
Delivery time:1-5 days after payment received.
|1||Main Instrument||1||Power Line|
|1||Operating Manual||1||USB Cable|
|1||Driving Software||1||Black/White Calibration Tiles|
|1||Color QC Software||1||Packing List|
Our CHNSpec Technology., Ltd are specialized on manufacturing spectrophotometers, colorimeters and gloss meters. Our products have gotten 10 Invention Patents including 1 American Invention
Patent. If any inquiry about our products, please contact me
freely. You are warmly welcome to visit and purchase from us.
Contact Name:Ms. Ivy Zhang
Tel:+86 (0)571 85888707
Fax:+86 (0)571 85888727
What's App:+86 15137145403
Q: Are you trading company or manufacturer ?
A: We are factory located in Hangzhou city. Welcome to come to
visit and purchase from us.
Q: How long is your delivery time?
A: Generally it is 1-3 working days if the goods are in stock.If
not, we will need 3-15 working days to prepare the products.
Q: Do you provide samples ? is it free or extra ?
A: Yes, and customer need to pay for the freight.
Q: What is your terms of payment ?
A: Payment is 100% T/T in advance.
Q: What's your MOQ?
A:Our MOQ is 1 set. If you order more than 5 sets, we will give