A Primer on pH
The term pH refers to the concentration of hydrogen ions (also called protons or H+) in a solution. Technically speaking, pH reflects the H+ activity, or amount of H+ available for reactions; but for many ordinary solutions, this value is very close to the total H+ concentration. For water solutions, the scale ranges from 0 to 14, with pure water right in the middle at 7. The more acid a solution is, the lower the pH reading, and alkaline solutions come in at the high end of the scale.
This might seem backwards, since acid solutions have the highest H+ content, but this shorthand notation keeps the math simple. For example, the H+ concentration in pure water is 1.0 x 10–7 moles/liter (10–7 molar, or M), so the pH is 7. If you put enough vinegar in the water to increase the H+ concentration to 10 times the original amount, or 10–6 M, the pH is 6. Likewise, you can add enough sodium hydroxide to pure water to neutralize 90% of the available H+ (making it 10–8 M in H+), and the pH value rises to 8. Smaller changes in the H+ concentration are expressed as decimal values: 6.3 or 7.5, for example.
Paper test strips are good for measuring approximate pH values, but when requiring something more exact—an instrument called a pH meter that can pick up even small changes in acidity. A pH meter is a boxy-looking instrument attached to a glass or plastic tube called a probe. (A bench top model is shown above. Handheld pH meters have a probe directly attached to the instrument body.) The probe has a glass bulb on one end and an electrical wire on the other. The wire sends data to the instrument when the glass bulb is dipped into a sample solution.
The pH meter measures H+ concentration by sensing differences in the electrical charges inside and outside of the probe. The glass bulb is made from silica (SiO2) that contains added metal ions. Most of the oxygen atoms in the glass are surrounded by silicon and metal atoms. However, the oxygen atoms on the inside and outside surfaces of the bulb are not completely surrounded, and they can “grab” positively charged ions from the solution.
When the bulb is dipped into an acid solution, H+ ions bond with the outside surface of the glass bulb, forming electrically neutral Si–OH groups. The Si–O– groups on the inside surface are in contact with a reference solution. The difference in electrical charge between the two surfaces creates an electrical potential, or voltage, and this causes an electrical current to flow through the wire at the other end of the probe.
Alkaline solutions have low concentrations of H+ ions and higher concentrations of negative ions such as OH–. The excess negative charges are balanced with positively charged metal ions such as Na+, and these positive ions hover close to the surface of the bulb rather than binding to the Si–O– groups. This sets up a different sort of charge separation, and the resulting electrical signal registers a high pH.
Specialized probes can measure concentrations of ionic species other than H+, including fluoride, chloride, ammonia, sodium, potassium, calcium, sulphide, and nitrate. In general, these probes are referred to as “specific-ion electrodes”.
This article first appeared on July 12, 2004.
Our Sulzer Cartridge Sprayer We now offer a compact, low cost dual cartridge sprayer for use with all ARCOR Coating products. We see it as a tool for smaller jobs where a big sprayer is not feasible. We use a small electric compressor from Home Depot that puts out 125psi, 5.7 CFM (of course any
Deairator Tank Repair DOWNLOAD PDF Case History #021 After many years of service, deairator tank needed to be restored due to severe metal loss caused by corrosion. Surfaces were White Metal-blasted to remove all pitted corrosion. Primer coat of ARCOR™ S-20 White was hand-applied. An intermediate coat of ARCOR™ S-20 Fuchsia was then applied, followed by
TDS SDS Data Sheet View Application Sheet Data Sheet GENERIC TYPE: AMINE CURED 100% SOLIDS EPOXY / URETHANE TOPCOAT DESCRIPTION AND RECOMMENDED USES: ARCOR™ SPRAYTHANE FL is a high elongation, elastomeric, hybridized epoxy/urethane. It is designed as a high resilience, high tear resistant, high impact resistant coating that is ideal to combat cavitation and abrasion, and it
Pulp & Paper Systems Protection MillGuard epoxy based coating systems are used to restore and protect equipment used in a wide variety of high-service paper mill applications. MillGuard Systems have become a valuable resource for maintenance managers by reducing downtime and extending the service life of system components. MillGuard Systems from ARCOR: 100% Solids Formulations,
Waterbox Restoration DOWNLOAD PDF Case History #026 REV.04-97 ARCOR™ protective coating system was installed at the manufacturer’s facility. Waterbox was new and specified ARCOR™ coatings prior to being put in service. Entire waterbox was abrasive-blasted to White Metal (SSPC-10) and then primed with ARCOR™ S-30 Prime. A layer of ARCOR™ S-30 Blue was then applied
Elcometer Pull Tester: To use pull test you have to ‘glue’ down the dolly’s. You sand 1” circles, wipe with Acetone. Mix some ARCOR 25551 from tubes kits. Apply to dolly & set dolly onto prepped area. To get a good reading you will have to let cure for at least a couple of hours.