  Dynamic phase microscopy has been applied for measurements of nanometer-scale
displacements of a piezoelectric scanner. This scanner, which was designed for
calibration purposes for scanning probe microscopy and TEM, exhibited a linear
and hysteresis-free translation in the 0.05-20 nm range. The voltage-activated
motion is described by a coefficient of 0.03 \pm 0.005 nm/V.
