  A scanning force microscope was implemented operating at temperatures below
4.2K and in magnetic fields up to 8T. Piezoelectric quartz tuning forks were
employed for non optical tip-sample distance control in the dynamic operation
mode. Fast response was achieved by using a phase-locked loop for driving the
mechanical oscillator. Possible applications of this setup for various scanning
probe techniques are discussed.
