  We constructed a dilution-refrigerator (DR) based ultra-low temperature
scanning tunneling microscope (ULT-STM) which works at temperatures down to 30
mK, in magnetic fields up to 6 T and in ultrahigh vacuum (UHV). Besides these
extreme operation conditions, this STM has several unique features not
available in other DR based ULT-STMs. One can load STM tips as well as samples
with clean surfaces prepared in a UHV environment to an STM head keeping low
temperature and UHV conditions. After then, the system can be cooled back to
near the base temperature within 3 hours. Due to these capabilities, it has a
variety of applications not only for cleavable materials but also for almost
all conducting materials. The present ULT-STM has also an exceptionally high
stability in the presence of magnetic field and even during field sweep. We
describe details of its design, performance and applications for low
temperature physics.
