  Further achievements of the XMM-Newton cross-calibration - XMM internal as
well as with other X-ray missions - are presented. We explain the major changes
in the new version SASv6.5 of the XMM-Newton science analysis system. The
current status of the cross-calibration of the three EPIC cameras is shown.
Using a large sample of blazars, the pn energy redistribution at low energy
could be further calibrated, correcting the overestimation of fluxes in the
lowest energy regime. In the central CCDs of the MOSs, patches were identified
at the bore-sight positions, leading to an underestimation of the low energy
fluxes. The further improvement in the understanding of the cameras resulted in
a good agreement of the EPIC instruments down to lowest energies. The latest
release of the SAS software package already includes corrections for both
effects as shown in several examples of different types of sources. Finally the
XMM internal cross-calibration is completed by the presentation of the current
cross-calibration status between EPIC and RGS instruments. Major efforts have
been made in cross-calibrations with other X-ray missions, most importantly
with Chandra, of course, but also with currently observing satellites like
Swift.
