  At National Institute of Metrological Research (INRIM), an automated setup to
calibrate DC Voltage generators, mainly top-level calibrators from 1 mV to 1 kV
has been developed. The heart of the setup is an INRIM-built automated fixed
ratios DC Voltage divider. The significant achievement of this setup is the
possibility to interconnect the divider, a DMM characterized in linearity, a DC
Voltage Standard and a DC Voltage generator under calibration and automatically
to manage the calibration process. This calibration method allows to save a lot
of time, to improve the reliability and to increase the accuracy of the
calibration of generators. The relative uncertainties of the system span from
0.6x10-6 to 1.2x10-4 improving the previous capabilities of the INRIM
laboratory for calibration of programmable multifunction instruments. In
addition, this system allows to avoid the employment of several Standards (some
of them still manually operating) carrying out the entire process without
changing the setup configuration and without the presence of operators. The
concept of this setup can be transferred to secondary high-level electrical
calibration Laboratories that could be consider it useful for their calibration
activities.
