  We present the design and construction of a SQUID-based magnetometer for
operation down to temperatures T = 10 mK, while retaining the compatibility
with the sample holders typically used in commercial SQUID magnetometers. The
system is based on a dc-SQUID coupled to a second-order gradiometer. The sample
is placed inside the plastic mixing chamber of a dilution refrigerator and is
thermalized directly by the 3He flow. The movement though the pickup coils is
obtained by lifting the whole dilution refrigerator insert. A home-developed
software provides full automation and an easy user interface.
