  A removable holder carrying a set of miniature temperature sensors inside our
Rayleigh-B\'enard convection cylindrical cell has been developed. The sensors
should not disturb the temperature and velocity fields of the buoyancy-driven
turbulent flow of our working fluid - cryogenic helium gas (~ 5 K). The cell,
300 mm both in diameter and height, must withstand the overpressure up to 2.5
bar, which is why all its parts are welded together. The holder consists of a
pair of identical feedthroughs (each with 6 glass-sealed pins), constantan
wires (0.1 mm), Be bronze springs and insulating elements and enables
installing up to 6 sensors simultaneously, with the option to replace the
damaged ones. We already used two such holders and obtained a map of local
temperature and its fluctuations in 12 positions inside the cell.
