  Purpose: 4D CBCT is a beneficial tool for the treatment of movable tumors,
because it can help us to understand where the tumors are actually located and
have a precise treatment plan. However, there is a limitation that general CBCT
images cannot perfectly help the sophisticated registration. On the other hand,
SymmetryTM 4D IGRT system of Elekta can offer the 4D CBCT registration option.
In this study, we intend to evaluate the usefulness of SymmetryTM. Method and
Materials: Planning CT images of the CIRS moving lung phantom were acquired
from 4D MDCT. And they are sorted as 10 phases from 0% phase to 90% phase. The
thickness of CT images was 1 mm. Acquired MDCT images were transferred to the
contouring software and a virtual target was generated. An one arc VMAT plan
was performed by using the treatment planning system on the virtual target.
Finally, the movement of the phantom was verified through XVI SymmetryTM
system. Results: The physical movement of CIRS moving lung phantom was +/- 10.0
mm in superior-inferior direction, +/- 1.0 mm in lateral direction, and +/- 2.5
mm in anterior-posterior direction. The movement of the phantom was measured
from 4D MDCT registration as +/- 10.2 mm in superior-inferior direction, +/-
0.9 mm in lateral direction, and +/- 2.45 mm in anterior-posterior direction.
The movement of the phantom was measured from SymmetryTM registration as +/-
10.1 mm in superior-inferior direction, +/- 0.9 mm in lateral direction, and
+/- 2.4 mm in anterior/posterior direction. Conclusion: It is confirmed that 4D
CBCT is a beneficial tool for the treatment of movable tumors. Therefore, 4D
registration of SymmetryTM can increase the precision of the registration when
a movable tumor is a target of radiation treatment.
