  In this paper, we have designed a low-cost scanning tunneling microscope
(STM) priced at 300 USD or 2000 CNY. This microscope is suitable for
educational purposes and low-demand research imaging at the nanometer level.
This microscope's motion components and scanner are controlled using
piezoelectric materials, avoiding the thermal drift associated with traditional
motor control. Our tip approach algorithm, which considers the capacitance and
friction characteristics during piezoelectric slider movement, has reduced the
time required for sample loading to establish tunneling current to
approximately 1 minute. The dimensions of the microscope body are
45x45x31.5mm(WxLxH), and the control voltage does not exceed 15V, ensuring the
safety of operators with limited experience. In the performance verification,
we performed a scanning tunneling scan on a Highly Oriented Pyrolytic
Graphite(HOPG) sample with bias voltages of 50mV and 60mV, resulting in clear
observations of the atomic features of HOPG in the STM pattern.
