  The tracking system of the HERMES spectrometer behind the bending magnet
consists of two pairs of large planar 6-plane drift chambers. The design and
performance of these chambers is described. This description comprises details
on the mechanical and electronical design, information about the gas mixture
used and its properties, results on alignment, calibration, resolution, and
efficiencies, and a discussion of the experience gained through the first three
years of operation.
