  We describe the conceptual design of the camera cryostats, detectors, and
detector readout electronics for the SuMIRe Prime Focus Spectrograph (PFS)
being developed for the Subaru telescope. The SuMIRe PFS will consist of four
identical spectrographs, each receiving 600 fibers from a 2400 fiber robotic
positioner at the prime focus. Each spectrograph will have three channels
covering wavelength ranges 3800 {\AA} - 6700 {\AA}, 6500 {\AA} - 10000 {\AA},
and 9700 {\AA} - 13000 {\AA}, with the dispersed light being imaged in each
channel by a f/1.10 vacuum Schmidt camera. In the blue and red channels a pair
of Hamamatsu 2K x 4K edge-buttable CCDs with 15 um pixels are used to form a 4K
x 4K array. For the IR channel, the new Teledyne 4K x 4K, 15 um pixel,
mercury-cadmium-telluride sensor with substrate removed for short-wavelength
response and a 1.7 um cutoff will be used. Identical detector geometry and a
nearly identical optical design allow for a common cryostat design with the
only notable difference being the need for a cold radiation shield in the IR
camera to mitigate thermal background. This paper describes the details of the
cryostat design and cooling scheme, relevant thermal considerations and
analysis, and discusses the detectors and detector readout electronics.
