  Absolute calibration of the Pierre Auger Observatory fluorescence detectors
uses a light source at the telescope aperture. The technique accounts for the
ombined effects of all detector components in a single measurement. The
calibrated 2.5 m diameter light source fills the aperture, providing uniform
illumination to each pixel. The known flux from the light source and the
response of the acquisition system give the required calibration for each
pixel. In the lab, light source uniformity is studied using CCD images and the
intensity is measured relative to NIST-calibrated photodiodes. Overall
uncertainties are presently 12%, and are dominated by systematics.
