TEST-902 2000A Large current injection generatorProduct DescriptionTEST-902 primary current injection test set use ARM chip to control the output process and large capacity of toroidal transformer, and equipped with LCD screen display current meter, also showed primary current value, second current value, turn ratio and the action time. Equipped with aluminum alloy chassis, PC panel, which has the advantages of beautiful appearance, large output capacity, small volume, light weight etc...It is mainly applied to test CT turn ratio in the power system and the contact resistance etc, which need large current.SpecificationsElectrical parametersPrimary currentSecondary measurementCurrent durationMechanical parametersEnvironmental conditions

Accuracy

±0.5%

Power Supply (AC)

220V±10%·50A, 50Hz/60Hz or110V±10%·100A, 50Hz/60Hz

Current Output (AC)

2000A·6V or 1000A·12V

Capacity

12KVA

The current output waveform distortion

≤5%

Range

0-2200A

Resolution

0.1A

Accuracy

<0.5%

Range

0-6A

Resolution

0.001A

Accuracy

<0.5%

2000A range

2000A 2-3minutes1000A 3-6minutes

1000A range

1000A 2-3minutes

Dimensions (W×D×H) (mm)

360×300×295

Weight (kg)

Host 35kg,External voltage regulator weight 15kg

Temperature

-10°C to 40°C

Relative humidity

≤80%

Primary Current Injection Test Set max current output 2200A, it can test current transformer turn ratio, and as large current generator.