The Health and Safety Executive (HSE) has become aware of two suppliers of scientific equipment that have supplied mesh gauzes with asbestos-containing centres. The material on the gauzes is used for its heat-resistant properties. The gauzes are conventionally used with tripods and Bunsen burners as shown in the photograph.
Although steps have been taken to prevent further supply by the suppliers identified so far, this will have implications for schools, colleges and other facilities which use these mesh gauzes.
Any gauzes that contain asbestos should not be used, and they must be safely disposed of as asbestos waste. As a precaution, if you are unclear as to what any gauze coating is made of then you should assume that it contains asbestos and follow the instructions below.
Analysis of the gauze material by HSE has identified that it is fibrous in nature and contains some asbestos, which is a known hazardous substance. Any risk from asbestos depends on the extent of asbestos fibre release and inhalation of these very fine fibres.
As a result, any free fibre release into the air will be minimal for normal use. However, the material is soft and crumbly and some small particles or fragments may detach on occasions including during use. Particles and debris may also break off over time through abrasion or impact in storage. These particles and fragments do not represent an airborne risk.
As a precautionary measure, and in line with the general legal requirement to prevent exposure to asbestos, the gauzes should be disposed of (see below). Any debris in containers or boxes should be carefully cleaned up and also disposed of.
Although suppliers identified so far will be contacting everyone they have supplied, it is advised that schools should initially adopt a precautionary approach as gauzes containing asbestos cannot be easily distinguished from those that do not. As such, teaching establishments should not handle, use or move their current stock of wire/mesh gauzes until they have checked with their supplier to find out whether they are likely to contain asbestos.
Action will need to be taken where asbestos content is confirmed or cannot be ruled out. The gauzes must be taken out of use and disposed of in accordance with relevant published guidance (see below).
Where non-asbestos-containing gauzes are stored directly alongside asbestos-containing gauzes, these should be treated as contaminated waste. Other equipment (such as clay triangles, Bunsen burners etc) which has been stored with asbestos-containing gauzes can be wiped clean with a damp cloth if there is any visible dust present. The cloth should be disposed of as contaminated waste.
The work to dispose of asbestos-containing gauzes and waste items is a low-risk activity, but it still needs precautions and controls in line with legal requirements. The work should only be done by people who are confident that they can follow the relevant guidance and have access to the right equipment. Assistance can be obtained from a specialist asbestos contractor (eg a contractor holding an asbestos work licence from HSE) but this is not a legal requirement in this situation.
Where possible, the gauzes stored in a container should be disposed of in that container to prevent further handling of individual gauzes. The container and gauzes should be treated as asbestos-containing waste.
Where this is not possible, gauzes should be carefully wetted using a hand-held spray bottle containing water with a small quantity of detergent (eg washing-up liquid) and handled carefully to prevent any further damage. They should be placed in a suitable heavy-duty polythene waste bag which is then placed in a second bag (ie double bagged) and labelled accordingly, in line with HSE's Asbestos Essentials Guidance EM9 Disposal of asbestos waste.
Use caution as the corners of the gauze may be sharp and could penetrate the polythene, so it may be more suitable to place the gauzes into a container such as a rigid, sealable plastic container before placing into suitable waste bags.
Any excess water (from spray) and dust/debris from the gauzes should be wiped up using a damp rag and the rag should be disposed of in the same manner, in line with the Asbestos Essentials guidance EM7 Using damp rags to clean surfaces of minor asbestos contamination.
As this is low-risk and short-duration work, respiratory protective equipment (RPE), ie a mask or respirator, is not legally required. However, dutyholders may wish to adopt a precautionary approach regarding the use of RPE and personal protective clothing. Where gauzes are damaged and/or there is resultant debris, a disposable coverall and a suitable dust mask (eg FFP3 or half mask with a P3 filter) could be worn. If a mask is used, the person should be face-fit tested and trained in its use. Full details are in Asbestos Essentials Guidance EM6 Personal protective equipment (including RPE).
When sourcing replacement gauzes, buyers should seek confirmation from their suppliers that the supply chain is assured and that new gauzes do not contain any level of asbestos.
The only certain way of assuring this is for items originating outside the EU to be analysed by a laboratory accredited for asbestos identification by the UK Accreditation Service (UKAS). You should seek a copy of a valid UKAS test certificate for the gauze from your supplier.
Our answers to frequently asked questions provide further information.